The combination of a thermogravimetric analyzer (TGA) with mass spectrometry (MS) to analyze evolved gases is a well-known technique. However, TG–MS does not allow differentiation between gases that evolve simultaneously; TG combined with gas chromatography–mass spectrometry (TG–GC–MS) can allow for a more complete characterization.
The use of a TGA also allows characterization of materials not suitable for gas chromatography–mass spectrometry (GC–MS) including solid residues left after pyrolysis. For example, fillers such as carbon black, glass fibres, or oxides; or the heavy oils remaining after a simulated distillation.Researchers can also produce heat capacity and enthalpy values for a reaction that is evolving gas by using a simultaneous differential thermal analyzer (STA) to introduce a sample to a gas chromatograph system. This can be extremely useful in combustion reactions such as the burning of a biofuel or in the release of a chemical during a condensation polymerization reaction.
Using TG to Understand the Leaching Process
GC–MS can differentiate between similar compounds, such as methanol and ethanol, and can also characterize complex gaseous mixtures coming off a sample. GC–MS can be used to detect extremely small levels of contaminants such as those in leachates. In fact, information can be collected on the small amounts of material evolved from complex matrices and this ability to analyze very small amounts of evolved gases is needed to understand processes like leaching, where a packaging material and/or its contents contaminate each other and degrade properties. This is a concern for many people as bisphenol A (BPA) and other plastizers can be leached into food. In some cases, researchers can even quantify the size of the effect by determining the amount of leachable material absorbed by using gas sample capture loops. This is often semi-quantitative as precise determinations can be difficult.
In the following two case studies, we look at the benefits of combining the techniques.
Case Study: Analysis of Coffee Beans in Plastic Containers
TG combined with GC–MS can be applied to the detection of leachable substances that contaminate foods and pharmaceuticals and which are often present at very low levels and in complex matrices, such as plasticizers or other packaging contaminants. It is important to understand the relationship of temperature with the evolution of the gas, to avoid misidentifying the products of decomposition or pyrolysis of the sample as a contaminant.
Method: The analysis was performed on a Pyris 1 TGA (PerkinElmer) using alumina pans and a standard furnace. The instrument was calibrated with nickel and iron and all samples were run under helium purge. Heating rates varied from 5 °C/min to 40 °C/min, depending on the sample under test. The furnace was burned off between runs in air. Samples were approximately 10–15 mg. Data analysis was performed using Pyris 9.0 Software (PerkinElmer). During the TG–GC–MS analysis, a Clarus 680 C GC–MS (PerkinElmer) was used. A 0.32 mm i.d. deactivated fused-silica transfer line was connected to the GC injector port. The transfer line was heated to 210 °C and connected to a 30 m × 0.32 mm, 1-µm Elite WAX column (PerkinElmer). In both cases, data analysis was performed using TurboMass GC–MS software (PerkinElmer).