CUSTODION™ SPME Syringes provide a simple, low cost methodology to collect and introduce samples to an analytical instrument.
The low cost and convenience of SPME allows multiple samplers to be deployed simultaneously. When coupled with the TRIDION™-9
GC–MS samples covering a wide boiling point range can be analyzed in as little as 3 min per run.
Sample concentrations can vary greatly from locations in or near industrial environments to surrounding residential environments
that are in close proximity. SPME samplers can be setup to cover a wide area, and sampling times adjusted based on anticipated
concentrations. In this application the detection capabilities of the TRIDION-9 and SPME are demonstrated for a number of
analytes at approximately 1 part per billion (ppb) v/v in air.
A 10 L Tedlar™ bag was spiked with a mixture of volatile organic compounds at ~1 ppb. Concentrations were corrected for atmospheric
pressure and temperature using US EPA method 10-2.4. The analytes were extracted from the gas phase at ambient temperature
for 60 min using a CUSTODION™ SPME syringe with a 65 µm polydimethylsiloxane/divinylbenzene (PDMS/DVB) fiber.
Following sample extraction, the SPME syringe was inserted into the TRIDION-9 GC-TMS injection port where the analytes were
desorbed into a split-splitless injector (270 °C) coupled with a low thermal mass, capillary GC column (MXT-5, 5 m × 0.1 mm,
0.4 µm df). After an initial 10 s hold at 50 °C, the GC temperature was increased at 2 °C/s to 270 °C for a total run time
of 1 min 30 s. The capillary GC is coupled to a TMS detector having a mass range of 43–500 m/z.
Figure 1 shows the GC-TMS analysis of the ~1 ppb sample, illustrating the total ion chromatogram and individual ion traces
for the extracted target compounds. Even with what is supposed to be a relatively clean sample, like one prepared in a Tedlar™
bag, detection and identification at trace levels can be problematic. Tedlar™ bags contain two major known contaminates N,N-Dimethylacetamide
and Phenol. The target compounds were automatically identified by deconvolution software in the presence of these large interfering
Figure 1: Total ion chromatogram, extracted ion chromatograms, and results table.
The CUSTODION SPME Syringe and TRIDION-9 GC-TMS are uniquely suited for rapid onsite analysis. The ease of use, low cost of
SPME and the high speed analysis of the TRIDION-9 GC-TMS allows for an unprecedented number of samples to be processed. This
high speed analytical capability provides the information when and where it is needed to make in-field decisions. Wide concentration
ranges can be accommodated with SPME by simply adjusting the sampling time.
Torion® , TRIDION™ and CUSTODION™ are trademarks of Torion Technologies Inc. The CUSTODION SPME Syringes are manufactured
and sold under license from SUPELCO under U.S. Patent 5,691,206, and/or any divisions, continuations, or revisions thereof.
Torion Technologies Inc.
796 East Utah Valley Drive, Suite 200, American Fork, UT 84003
tel. (801) 705-6600, fax (801) 705-6649