LC Trouble Everywhere - - Chromatography Online
 Chromatography Online All results
LC Trouble Everywhere
 Jan 1, 2012 LCGC North America Volume 30, Issue 1, pp. 36-41
 No matter where liquid chromatography is used, it is never immune to problems. One of the privileges of teaching liquid chromatography (LC) classes for a living is that I get to meet chromatographers from around the world during my travels. As I write this installment, I'm just finishing six weeks of classes, where I have had a chance to discuss problems that attendees were having with a variety of LC applications. Although the topics in this column originate in Poland, Turkey, and Argentina, they are no different than those that arise in Singapore, Galway, Charlottetown, or Huntsville. Let's look at a couple of problems that students had related to both conventional LC and ultrahigh-pressure LC (UHPLC). These problems apply equally to both techniques. How Much Can I Inject? One of my students from Poland, who was just starting out using UHPLC, asked if she could inject 5 µL of sample onto her UHPLC column without problems. Her conditions comprised a 75 mm × 2.1 mm column packed with 1.8-µm diameter particles operated at a flow rate of 0.8 mL/min. It is fairly easy to perform a few simple calculations and answer her question. A good rule of thumb (1) is that you can inject ≈15% of the volume of the first peak of interest without causing a more than ≈5% increase in peak broadening, as long as you use mobile phase for the injection solvent. We need to determine the volume of the first peak of interest to see how this applies in the present case. We will assume a retention factor, k, of 1 for the first peak; you can make similar calculations for other retention factors. First, we need to estimate a column plate number (efficiency), N, which can be done as N = 300 L/d p                        [1] where L is the column length in millimeters and d p is the particle diameter in micrometers. A well-packed column tested with an ideal solute will give larger values of N, but equation 1 is a reasonable estimate for real samples. With our L = 75 mm, d p = 1.8 µm column, we get N = 12,500. Next, it is necessary to estimate the column volume, V M, in milliliters, using the equation V M ≈ 5 × 10-4 Ld c 2                [2] where dc is the column diameter in millimeters. For the 75 mm × 2.1 mm column we get 0.165 mL. From this we can get the column dead time, t 0, by dividing by the flow rate, 0.8 mL/min: 0.21 min. (Note: If you are trying to repeat these calculations, carry along several extra decimal places of precision — I'm rounding the numbers for simplicity of presentation). With the knowledge of the column dead time, we can calculate the retention time, t R, for our k = 1 peak by rearranging the equation for k: t R = t 0 (1 + k)                     [3] This gives t R = 0.41 min. The next to last step is to estimate the peak volume of the first peak. We estimated N with equation 1, but if we were measuring N from an isocratic chromatogram, we would use N = 16 (t R/w)2                   [4] where w is the baseline width of the peak measured between tangents drawn along the sides of the peak. Solving equation 4 for the peak width, we get w = 4 t R/N 0.5                     [5] For t R = 0.41 min and N = 12,500, this gives w = 0.015 min, or at 0.8 mL/min, w = 11.8 µL. Using our 15% rule of thumb from the beginning of our discussion, an injection of (0.15 × 11.8 µL) = 1.8 µL, or ≈ 2 µL would be allowed in the mobile phase. This is less than half the desired 5-µL injection, so it is expected that excess peak broadening would be observed with a 5-µL injection. Armed with this knowledge, I would perform a series of injections of 1, 2, 3, 4, and 5 µL and observe the effect on the peak width of the first peak of interest. I agree wholeheartedly with Izaak Kolthoff, the father of analytical chemistry, who is credited with the quote, "Theory guides, experiment decides." So, use the results of the injection-volume experiment for the final decision.

 Survey
Column technology
In what direction do you see HPLC column technology going in the next five years?
Smaller-diameter particle development
More unique stationary phases
Improvements in design/better integration into the hardware
A consolidation of recent column developments
Smaller-diameter particle development 13%
More unique stationary phases 48%
Improvements in design/better integration into the hardware 13%
A consolidation of recent column developments 26%

LCGC COLUMNISTS 2013

 Column Watch | Ronald E. Majors: LCGC Columnist Ron Majors, established authority on new column technologies, keeps readers up-to-date with new sample preparation trends in all branches of chromatography and reviews developments in existing technology lines. LATEST: Application of Existing Sample Preparation Technologies to New Areas History of Chromatography | Industry Veterans: With each installment of this column, a different industry veteran covers an aspect of the evolution and continued development of the science of chromatography, from its birth to its eventual growth into the high-powered industry we see today.LATEST: Professor Barry L. Karger: Scientist, Mentor, and Innovator MS — The Practical Art | Kate Yu: Kate Yu is the editor of 'MS-The Practical Art' bringing her expertise in the field of mass spectrometry and hyphenated techniques to the pages of LCGC. In this column she examines the mass spectrometric side of coupled liquid and gas-phase systems. Troubleshooting-style articles provide readers with invaluable advice for getting the most from their mass spectrometers. LC Troubleshooting | John Dolan: LC Troubleshooting sets about making HPLC methods easier to master. By covering the basics of liquid chromatography separations and instrumentation, John Dolan, Vice President of LC Resources and world renowned expert on HPLC, is able to highlight common problems and provide remedies for them. LATEST: What's the Problem with the LLOQ — A Case Study
More LCGC Chromatography-Related Columnists>>
Multimedia
 Editors Series' : Integrating On-Line Sample Preparation and HPLC Separations with Triple Quadrupole Mass Spectrometry for Ultratrace Quantitative Analysis of Small Molecules from Complex Matrices October 30, 2013
 Maintain your Methods, Prolong your Helium Supply, and Save Your Budget with an Innovative Helium Solution October 30, 2013