Underground storage tanks (USTs) for various petroleum-based substances such as gasoline, diesel fuel and fuel oil are monitored
for leaks as authorized by the Resource Conservation and Recovery Act (RCRA). Leaking USTs (LUSTs) contaminate ground water
and soil. Besides the environmental impact, there are significant financial consequences associated with fines and cleanup.
UST monitoring is usually done by gas chromatography with flame ionization detection (GC–FID). Because these petrochemical
samples often have a wide molecular weight range, a temperature-programmed run is required to get adequate retention of higher
volatility components and subsequently elute the high molecular weight components in the same chromatographic run. The chromatographic
run time of these types of samples typically takes more than 20 min. Thermo Scientific UltraFast technology makes it possible
to reduce the total analysis time to less than four min. A Total Recoverable Petroleum Hydrocarbons (TRPH) analytical standard
mixture is representative of the molecular weight range typically found in UST samples.
Figure 1: Top: Using UltraFast GC. Bottom: Using conventional GC.
Thermo Scientific UltraFast TRACE GC Ultra Configuration
Column: UltraFast UFC-M1 Phase GC Column, 2.5 m × 0.1 mm × 0.4 µm
Oven: 50 °C (0.5 min hold) -340 °C (2 min hold) at 200 °C/min
Helium carrier gas at 0.5 mL/min
Thermo Scientific UltraFast column technology can be used to achieve a fast analysis of a wide molecular weight range of petroleum
hydrocarbons such as those found in UST monitoring. Excellent separation of a mixture of even-numbered alkanes from C8 to C40 was accomplished in less than 3.5 min. Analysis of these types of samples on a conventional GC capillary column requires >
20 min. Since temperature programming is required for this analysis, the time needed for the GC oven to cool down to the initial
temperature also needs to be considered. The time required to cool down the UltraFast from 340 °C to 50 °C for this analysis
is approximately 90 s, where the time required to cool down a conventional GC oven for the same temperature range is approximately
4 min, thus further decreasing the overall run-to-run time. Laboratories doing UST analysis can do approximately five times
more analyses per day using UltraFast technology vs. conventional GC columns.
Analysis of sample containing a wide molecular weight range of hydrocarbons such as those found in Underground Storage Tanks
(UST) monitoring can be done in slightly more than three min with the use of a Thermo Scientific UltraFast column. This is
approximately five times faster than doing the same analysis using conventional GC capillary columns.
1. Underground Storage Tanks, Office of Solid Waste and Emergency Response (OSWER), U.S, EPA. http://www.epa.gov/oust/ (accessed June 12, 2012).
2. Leaking Underground Storage Tank Program Implements The Recovery Act, EPA Information Related to the American Recovery
and Reinvestment Act of 2009 (Recovery Act), U.S., EPA.
http://www.epa.gov/oust/eparecovery/ (accessed June 12, 2012).
3. Storage Tank Regulation, Florida Department of Environmental Protection.
http://www.floridadep.org/waste/categories/tanks/ (accessed June 12, 2012).
Receive the complete application note at:
Thermo Fisher Scientific Inc.
320 Rolling Ridge Drive, Bellefonte, Pennsylvania 16823, USA
tel. (800) 532 4752