LC Trouble Everywhere - - Chromatography Online
 Chromatography Online All results
LC Trouble Everywhere
 Jan 1, 2012 LCGC North America Volume 30, Issue 1, pp. 36-41

Reference Standard Purity

 Figure 1: Reconstructed chromatogram for a certificate of analysis. The arrow marks the column dead time.
In Turkey, a student complained that he had purchased a reference standard that was claimed to be >99% pure in the certificate of analysis, but when he ran it on his LC system, he found impurity peaks that greatly exceeded the <1% claim. He showed me a chromatogram that he obtained with a large analyte peak and a smaller impurity peak that indeed was more than 1% of the reference standard. He also had a copy of the certificate of analysis that contained a chromatogram showing a single peak that looked similar to the one I've recreated in Figure 1. The mobile phase was 90% acetonitrile, 10% water on a C18 column for the certificate of analysis; unfortunately, I did not note the conditions for his analysis.

However, even with this limited amount of information, it is possible to make an educated guess about the source of the problem. If you recall from previous installments of this column, I've stated many times that the ideal isocratic chromatogram will have peaks in the 2 < k < 10 range, or if this is not possible, 1 < k < 20. This gives the analytes sufficient time to interact with the column to achieve "good" chromatography, yet keeps the run time from being excessive. Normally, we calculate the retention factor, k as

k = (t Rt 0)/t 0                   [6]

where t R and t 0 are the retention time and column dead time as noted in equation 3. I did not include a time axis in Figure 1, because I can't remember what it was in the certificate of analysis, but the units in equation 6 cancel, so we can use a ruler to calculate k from the chromatogram. The first disturbance in the baseline (arrow in Figure 1) marks t 0 and we can use the top of the peak as t R to calculate k ≈ 0.3. This is much less than the desired minimum k = 2. When peaks are eluted with k << 1, there is little time for the analyte to interact with the column, and the chances of having unresolved peaks present is increased. This is especially important when using a chromatogram to certify peak purity. The small tail on the peak may be peak tailing that is normal with many peaks or it may be a subtle indication that an impurity is present. There is no excuse for producing a chromatogram like this for a certificate of analysis, except laziness, ignorance, or impatience. It would be easy to increase k for the peak from 0.3 to 2 < k < 10 by changing to a weaker mobile phase, such as 70% acetonitrile, 30% water. At that point, a more convincing case could be made to show that the peak was indeed >99% pure. Although I don't remember the conditions of the user's chromatogram, I do remember that the analyte peak was significantly broader than that in the certificate of analysis, suggesting that the retention time was much larger. My answer to the question was that the certificate of analysis didn't convince me of the purity of the reference standard, so I would be more likely to trust the analyst's chromatogram in which k >> 0.3.

Conclusions

It doesn't matter where we live; LC problems have no international boundaries. In fact, I've noticed that chromatographic terminology often is adopted directly into many different languages. In some ways, chromatography really is a universal language. It reminds me of a time when a visitor from one of the Asian countries visited my son's classroom when my son was about 12 years old. The visitor asked if the kids wanted to learn some of his language. All were excited about the prospect. The visitor carefully recited, "computer," "Coca-Cola," and "McDonalds." Maybe he should have included "chromatography."

 John W. Dolan
John W. Dolan "LC Troubleshooting" Editor John Dolan has been writing "LC Troubleshooting" for LCGC for more than 25 years. One of the industry's most respected professionals, John is currently the Vice President of and a principal instructor for LC Resources in Walnut Creek, California. He is also a member of LCGC's editorial advisory board. Direct correspondence about this column via e-mail to
.

References

L.R. Snyder, J.J. Kirkland, and J.W. Dolan, Introduction to Modern Liquid Chromatography, 3rd Ed. (Wiley, Hoboken, New Jersey, 2010), p. 120.

 ADVERTISEMENT
blog comments powered by Disqus

Global E-newsletters subscribe here:

 Survey
CHROMacademy
Are you aware of CHROMacademy and the learning tools it offers?
Yes, I would like to know more
Yes, I am not interested
No, I would like to know more
No, I am not interested
Yes, I would like to know more 67%
Yes, I am not interested 9%
No, I would like to know more 19%
No, I am not interested 5%

LCGC COLUMNISTS 2014

 Sample Prep Perspectives | Ronald E. Majors: LCGC Columnist Ron Majors, established authority on new column technologies, keeps readers up-to-date with new sample preparation trends in all branches of chromatography and reviews developments in existing technology lines. LATEST: The Role of Selectivity in Extractions: A Case Study History of Chromatography | Industry Veterans: With each installment of this column, a different industry veteran covers an aspect of the evolution and continued development of the science of chromatography, from its birth to its eventual growth into the high-powered industry we see today. LATEST: Georges Guiochon: Separation Science Innovator MS — The Practical Art| Kate Yu: Kate Yu is the editor of 'MS-The Practical Art' bringing her expertise in the field of mass spectrometry and hyphenated techniques to the pages of LCGC. In this column she examines the mass spectrometric side of coupled liquid and gas-phase systems. Troubleshooting-style articles provide readers with invaluable advice for getting the most from their mass spectrometers. LC Troubleshooting | John Dolan: LC Troubleshooting sets about making HPLC methods easier to master. By covering the basics of liquid chromatography separations and instrumentation, John Dolan, Vice President of LC Resources and world renowned expert on HPLC, is able to highlight common problems and provide remedies for them. LATEST: LC Method Scaling, Part I: Isocratic Separations

LCGC North America Editorial Advisory Board>>

LCGC Europe Editorial Advisory Board>>

Multimedia
 Sample Preparation Method Development for Unconventional Matrices February 24, 2014
 Applying Best-in-Class HPLC Column Technologies to the Analysis of Proteins and Biotherapeutics March 11, 2014
 Editors' Series: Prospects and Pitfalls in Metabolomics: From Study Designs to Interpreting Data in Biomedical Applications March 11, 2014
 The Use of Solid Core Technology in the Pharmaceutical Environment January 23, 2014
 Increasing Laboratory Efficiency through a Harmonized Compliance Solution March 11, 2014
 VALIDATION RESOURCES FROM IVT NETWORK Process Validation Special Editions 19th Annual Validation Week Compendium Computer and Software Validation Volume II Special Edition Analytical Method Validation Toolkit More from IVT
Click here