Professor Mary Wirth and graduate students Bingchuan Wei and Benjamin Rogers from Purdue University demonstrate a quantum
leap in protein column efficiency. Using colloidal silica particles of submicrometer diameters (470 nm), they obtained plate
heights that were as much as 15fold lower than the theoretical limit for HagenPoiseuille flow. The smallest plate height
was 15 nm, 500fold smaller than previous reports for pressuredriven protein chromatography. This remarkable achievement
was attributed to "slip flow," which is explained in this installment. The authors also present a practical example using
this new concept.
The plate heights of commercial liquid chromatography (LC) columns have changed little over the last 20 years, following the
trend of H
_{min} ≈ 2d
_{p}, where d
_{p} is the particle diameter (1). Simulations have shown that H
_{min} ≈ d
_{p} is the limit for nonporous particles (2). With particle sizes dropping by only a factor of about two during the last 20 years,
as such plate heights have improved by the same factor. The reason is not a lack of diligence or creativity; instead, it is
because of the fundamental limits of mass transport, with the ultimate limit being the velocity spread of the mobile phase
across the spaces between particles. We recently used a nanofluidic phenomenon called slip flow (3–8), which greatly reduces this velocity spread, thereby reducing the plate height dramatically. We demonstrated a plate
height as low as 0.03 d
_{p}, which was 15 nm (9). This opens a new horizon for efficiency in ultrahighpressure liquid chromatography (UHPLC).
Figure 1: Illustration of how slip flow gives a narrower velocity distribution compared to Hagen Poiseuille flow. The velocity
is far from zero at the wall for slip flow.

The concept of slip flow is illustrated in Figure 1. Imagine that the walls are the surfaces of two adjacent particles. The
familiar HagenPoiseuille flow is depicted with its parabolic velocity profile, in which the velocity goes to zero at the
walls. This is the realm of chromatography today. The figure also depicts slip flow, in which the velocity does not go to
zero at the walls, thereby giving rise to a narrower distribution of velocities in the column. Slip flow occurs when the molecular
interactions between fluid and the surface are weak, allowing the fluid at the interface to have a nonzero velocity. The existence
of slip flow is common knowledge in the field of fluid dynamics, where it is typically a small correction factor when there
are weak fluid–surface interactions, but it becomes a big factor when the channel reaches the scale of tens of nanometers
(3–8). Consequently, slip flow always occurs in reversedphase LC, but it is not distinguishable from HagenPoiseuille flow
because the dimensions of chromatographic materials are relatively large. The nanoscale channel dimensions are achieved by
using submicrometer particles.

Experimentally, the parabolic flow profile cannot be directly viewed on the nanoscale, but it can be inferred because one
can measure a companion to the narrower flow profile: a faster flow rate. Because slip flow gives a nonzero velocity at the
wall, the volume flow rate, Q, is enhanced relative to that for HagenPoiseuille flow, Q
_{HP}. In equation 1, for a capillary of radius r, the term L
_{s} is the slip length, which is defined as the ratio of the velocity to shear rate, each at the wall.