The GC columns may be of two types: packed or capillary. Packed columns can accept a larger injection volume, but capillary columns have better separating power. There are a number of different detectors available as a function of the analyte constituent. These include thermal and electrolytic conductivity, flame ionization, electron capture, nitrogen–phosphorus, photoionization, mass selective, infrared, and atomic emission detectors.
GC is very well established in the U.S., Western Europe, and Japan, where the installed base is extremely large. In fact, North America and Europe combined account for nearly 60% of the worldwide GC demand. However, lower-end GC systems have become ubiquitous in areas such as Eastern Europe, China, India, and Latin America.The foregoing data were extracted and adapted from SDi's recently published 12th Edition. For more information, contact Glenn Cudiamat, VP of Research Services, Strategic Directions International, Inc., 6242 Westchester Parkway, Suite 100, Los Angeles, CA 90045, (310) 641-4982, fax: (310) 641-8851, e-mail: email@example.com