OR WAIT null SECS
Practical HPLC, HILIC and RP-LC methods of analysing common food and beverages containing polyphenols are presented.
The in-depth study of polyphenols is hampered by their sheer complexity, combined with the inherent limitations of conventional high performance liquid chromatography (HPLC) separation methods. In this paper, the beneficial application of a number of recent developments in HPLC for polyphenol analysis is highlighted. Small particle-packed columns operated at elevated pressures and temperatures are used to improve the speed and/or resolving power of conventional HPLC methods. In addition, comprehensive two-dimensional LC combining hydrophilic interaction chromatography (HILIC) and reversed phase LC (RP-LC) is shown to be a powerful technique to increase the resolving power. Examples of the successful application of these technologies for improved analysis of procyanidins, flavonols and anthocyanins in wine, tea, cocoa and blueberry samples are presented.
Polyphenols represent one of the most studied classes of natural compounds. Interest in these ubiquitous secondary metabolites, in recent times in particular, stems from their physiological contribution to the human diet and the numerous health benefits associated with their consumption. In addition, polyphenols are responsible for a range of organoleptic properties, including colour, astringency, bitterness and mouth-feel, as well as ageing properties.1
Clearly, the analysis of these compounds is of significant importance in various fields, particularly in the food and beverage and neutraceutical industries. However, the accurate analytical determination of phenolic compounds is hampered by the sheer complexity of the numerous chemical classes and species classified as polyphenols. These broadly include phenolic acids (hydroxybenzoic and cinnamic acids) and flavonoids, the latter class being subdivided into flavonols, flavanols (and their polymeric derivatives, the proanthocyanidins) and anthocyanins (anthocyanin-glycosides) (Figure 1).
Figure 1: General structures of the principal classes of phenolics. R is commonly H, OH or OCH3, while R1 may be OH, O-glycosyl, or O-acyl-glycosyl.
The investigation of polyphenols is therefore closely related to the development of suitable analytical techniques for their identification and quantification. High performance liquid chromatography (HPLC), and reversed phase LC (RP-LC) in particular, is most often used for the routine analysis of phenolics, while in recent years the hyphenation of HPLC to mass spectrometry (MS) has become an indispensable tool for the detailed investigation of these compounds.2
The resolving power of chromatographic methods is commonly measured in terms of peak capacity: the number of peaks that can theoretically be resolved with unit resolution using a given separation method. The peak capacity (np) for gradient separations can be calculated according to:3
where tg is the gradient run time and w the average (4σ) peak width. Common gradient RP-LC methods typically provide peak capacities in the order of 100–150 (for ~1 h analysis times). Considering that the peak capacity should significantly exceed the number of randomly distributed analytes in a given sample to provide a high likelihood of separation,4 and the fact that typical phenolic extracts contain in excess of 100 species, the limitations of HPLC are evident.
Importantly, in recent years a number of significant developments in HPLC have contributed to deliver improved performance. These include the use of ultra high pressure liquid chromatography (UHPLC), new developments in stationary phase morphologies (monolithic columns and superficially porous particles), high temperature liquid chromatography (HTLC) and multidimensional LC separations. In this contribution we discuss the beneficial application of some of these new technologies to improve the analysis of polyphenols in a variety of samples.
Reagents and materials: Cocoa beans, wine, blueberries and green tea were purchased from a local supermarket. Epicatechin, catechin and uracil standards were from Sigma-Aldrich (Atlasville, South Africa), and anthocyanin standards from Polyphenols Laboratories (Sandnes, Norway). Dimeric procyanidins were isolated by HILIC fractionation from a cocoa extract. Mobile phases were prepared from HPLC-grade solvents (Sigma) and Milli-Q deionized water (Millipore, Milford, Massachusetts, USA).
Sample preparation: Extracts of green tea and cocoa were prepared as reported previously.5,6 For extraction of blueberry anthocyanins, 6.681 g blueberries were frozen and ground under liquid nitrogen and extracted with methanol/water/formic acid (60:37:3), followed by evaporative pre-concentration. Filtered wine samples were directly injected.
Instrumentation: HPLC–UV analyses were performed on an Alliance 2690 HPLC equipped with a photodiode array (PDA) detector (Waters, Milford, Massachusetts, USA). UHPLC experiments were performed on an Acquity UPLC system with PDA detector (Waters). HPLC–ESI-MS analyses were performed on a UPLC system coupled to a Waters Ultima Q-TOF mass spectrometer operated in positive ionization (PI) mode.6 Blueberry anthocyanins were tentatively identified by MS and MS2 spectra and comparison of retention times with literature reports.7 The following columns were used: XBridge C18 (250 × 4.6 mm, 5 µm) and Acquity BEH C18 (50 and 100 × 2.1 mm, 1.7 µm, both Waters), Zorbax SB C18 (50 × 4.6 mm, 1.8 µm, Agilent, Waldbronn, Germany) and Develosil Diol-100 (250 × 1 mm, 5 µm, Nomura Chemical, Japan).
Cocoa procyanidins and green tea flavonols: HPLC analyses at 25 °C were performed using 0.1% formic acid (A) and acetonitrile (B) at a flow rate of 1 mL/min: 0–0.63 min (2%B), 0.63–27 min (2–18%B), 27–45 min (18–25%B), 45–51.75 min (25–100%B), 51.75–56.25 min (100%B). UHPLC analyses were performed on a 100 mm Acquity column at 50 °C using the following gradient at 0.3 mL/min: 0–0.17 min (2%B), 0.17–7.4 min (2–18%B), 7.4–12.48 min (18–25%B), 12.48–14.53 min (25–100%B), 14.53–15.6 min (100%B). UV detection was performed at 280 and 370 nm for procyanidins and flavonols, respectively.
Wine anthocyanins: Mobile phases consisted of 7.5% formic acid (A) and acetonitrile (B). HPLC analyses were performed on the XBridge column at 25 °C and 1 mL/min using the following gradient: 0–1 min (1%B), 1–12 min (1–13.5%B), 12–24 min (13.5–23.5%B), 24–28 min (23.5–28.5%B), 28–35 min (28.5%B). UHPLC analyses were performed on two coupled 100 mm Acquity columns at 50 °C and 0.06 mL/min as follows: 0–3 min (1%B), 3–34 min (1–13.5%B), 34–67 min (13.5–23.5%B), 67–78 min (23.5–28.5%B), 78–98 min (28.5%B). UV detection was performed at 500 nm.
Blueberry anthocyanins: Analyses were performed on the XBridge column using the same mobile phases as for wine anthocyanins. A flow rate of 1 mL/min was used throughout with the following gradients: 0–50 min (1–25%B), 50–55 min (25–40%B) at 25 °C; and 0–50 min (1–20%B), 50–55 min (25–40%B) at 70 °C. A post-column split (1:4) was used prior to eluent introduction into the MS.
HILIC×RP-LC: Conditions were as reported elsewhere.5,6
For procyanidins, experiments were performed on the XBridge column at 25 °C and on a 50 mm Acquity column at 25 °C and 50 °C, and for anthocyanins on the XBridge column at 25 °C and 70 °C. Mobile phases were as specified for each of these samples above, with the composition varied to ensure similar retention factors at all temperatures. Uracil was used as unretained marker, and all experiments were performed in duplicate. van Deemter curves and kinetic plots were constructed according to reference 8.
Uni-dimensional HPLC analyses: Since the inception of HPLC, there has been a continuous drive to improve the performance of the technique. The most widely practiced approach is to reduce the particle size (dp) of column packing. The benefit of this approach is illustrated by the van Deemter curve, which plots chromatographic efficiency, in terms of plate height (H = L/N, where N is the number of theoretical plates) as a function of linear velocity of the mobile phase, u0. The optimal linear velocity, uopt, is that mobile phase velocity (flow rate) which delivers the minimum plate height (Hmin) and therefore maximum efficiency. The effect of dp on the van Deemter curve is demonstrated in Figure 2(a) for a dimeric procyanidin analysed on 5 µm and 1.7 µm columns. It is clear that uopt increases and Hmin decreases with a reduction in dp, implying that faster and more efficient analyses are possible on the smaller phase for equal column lengths.
Figure 2: (a) Comparison of van Deemter curves for procyanidin dimer on 5 and 1.7 Âµm columns at 25 Â°C and 50 Â°C; (b) Corresponding kinetic plots of efficiency versus analysis time.
In fact, this picture is not entirely accurate, since decreasing dp leads to a very important decrease in column permeability. Pressure constraints ultimately limit the column length and therefore efficiency attainable, so that very high efficiencies are easier to achieve for larger particle sizes (a result of the higher permeability of the larger dp phases). Note though that the price to pay for such high efficiencies is a very significant increase in analysis time.
This dilemma has been partially overcome by the availability of a new generation of high-pressure instrumentation. In Figure 2(b) the isocratic efficiency as a function of total analysis time for the procyanidin dimer is compared for the same 5 µm and 1.7 µm columns, the latter operated at pressures up to 1000 bar. Note the significant decrease in analysis time achieved on the UHPLC column for an efficiency of 25000 plates (equal to that provided by a conventional 250 mm, 5 µm column). Moreover, UHPLC clearly offers faster analysis for efficiencies up to N ~80000.
It is relatively straightforward to practically exploit the benefits of UHPLC because HPLC methods may nowadays easily be adapted to UHPLC column formats and instrumentation, essentially providing the same separation mechanism. To date this approach has received comparatively little attention for polyphenol analysis,9,10 with the focus primarily on very fast analysis on short columns, essentially providing the same efficiency as conventional HPLC columns.
Increasing the temperature of HPLC separations provides an additional means of improving performance.11 This is primarily a result of faster analyte diffusion in the mobile phase at higher temperatures. The effect of this on plate height behaviour is demonstrated once again for the dimeric procyanidins on the 1.7 µm column in Figure 2: while the maximum column efficiency (= L/Hmin) remains roughly unchanged, the optimal linear velocity increases and efficiency-loss at high flow rates is reduced. At higher analysis temperature the flow rate should be increased to operate close to uopt, in this manner providing the same efficiency faster. Note that even though the mobile phase viscosity is inversely proportional to temperature, the requirement of operating at higher flow rates means that the overall pressure remains roughly constant.
Figure 3: Comparison of HPLC analysis of cocoa procyanidins and green tea flavonols at 25 Â°C on a 5 Âµm XBridge column (a) and (b) respectively, and at 50 Â°C on a 1.7 Âµm column (c) and (d) respectively. Detection was performed at 280 nm and 370 nm for procyanidins and flavonols, respectively. ÎPmax = 193 and 505 bar for HPLC and UHPLC analyses.
In fact, elevated temperature offers a complementary means of further increasing the speed of UHPLC separations for efficiencies up to N ~80000. The benefits of this approach are illustrated in Figure 3, where conventional HPLC analysis of cocoa procyanidins and green tea flavonols are compared to UHPLC analysis at 50 °C. This approach provides similar separation in a third of the analysis time, demonstrating obvious advantages for routine analysis of polyphenols.
Figure 4: van Deemter plots for four anthocyanins on a 5 Âµm XBridge column at (a) 25 Â°C and (b) 70 Â°C and (c) corresponding kinetic plots of efficiency versus retention time for malvidin-3-O-glucoside
For anthocyanins (anthocyanidin-glycosides, Figure 1), analysis temperature plays a particularly important role. These compounds are known to exist in solution in a number of different chemical forms.12 On-column inter-conversion between these species causes significant band broadening,13 leading to the relatively poor efficiency commonly observed for HPLC analyses of anthocyanins. Higher analysis temperatures speed up the inter-conversion reactions — leading to less band-broadening and improved efficiency. This is demonstrated for the van Deemter curves of four common anthocyanins in Figure 4. At conventional flow rates (i.e., 1 mL/min on 4.6 mm i.d. columns), an improvement in efficiency of between 25 and 130% is achieved at 70 °C compared with 25 °C. In this manner elevated temperature may be used to improve the efficiency of conventional gradient RP-LC analysis of anthocyanins, as demonstrated for blueberry anthocyanins in Figure 5. Also note that temperature may affect the selectivity of the separation, as is evident from the relative elution order of some anthocyanins in Figure 5.
Figure 5: Extracted ion chromatograms for the gradient HPLC-MS analysis of blueberry anthocyanins on a 5 Âµm XBridge column: (a) at 25 Â°C, and (b) at 70 Â°C, both at 1 mL/min. Cy = cyanidin, Dp = delphinidin, Pt = petunidin, Pe = peonidin and Mv = malvidin.
The previous examples demonstrate how analysis speed for polyphenol analysis may be increased (for the same efficiency) or efficiency increased for the same analysis time. Another approach especially suited for complex phenolic fractions, is to exploit the benefits of UHPLC and elevated temperature for high-efficiency analysis. This is demonstrated for the analysis of red wine anthocyanins on a 200 mm 1.7 µm column operated at 50 °C in Figure 6.14 This set-up provides isocratic efficiencies of up to 40000 plates, and gradient analysis delivers much improved resolution for an acceptable analysis time of 90 min. Note that this analysis is performed at pressures below 300 bar, implying that conventional instrumentation may be used (provided system delay and void volumes are optimized for such analyses).
Figure 6: Comparison of gradient RP-LC analysis of red wine anthocyanins on a (a) 5 Âµm XBridge column at 25 Â°C and 1 mL/min and a (b) 1.7 Âµm Acquity column at 50 Â°C and 0.06 mL/min.14 UV detection at 500 nm. ÎPmax = 190 and 295 bar for HPLC and UHPLC analyses.
However, caution should be exercised in the application of elevated temperature for phenolic analysis. As is well documented in literature, anthocyanins in particular are readily degraded at relatively mild temperatures. Fortunately this aspect may be evaluated quantitively. Using the approach of Thompson et al.15 it at can be demonstrated that analysis times of less than 2 hours at 70 °C do not lead to on-column degradation of anthocyanins.13
Finally, in the context of uni-dimensional HPLC analysis, it should be mentioned that a number of additional developments in column technology in recent years hold some promise for HPLC analysis of polyphenols. These include the use of monolithic16 and superficially porous particles.17 However, limited application of these column formats to polyphenol determination has been reported to date.18
The challenges associated with HPLC analysis of phenolics may be demonstrated using an example: procyanidins (Figure 1) consist of a mixture of monomeric, oligomeric and polymeric compounds composed of flavanol building blocks. As the degree of polymerization (DP) increases, the number of possible isomers increases exponentially (a result of the possible combinations of monomers as well as different inter-flavanol bonds). For example, 48 isomers of dimeric procyanidins containing catechin and epicatechin building blocks are possible.19 Considering that optimized high-efficiency HPLC methods provide peak capacities of maximally 400, no uni-dimensional HPLC method is capable of providing complete separation of higher molecular weight (MW) procyanidins in a single analysis.
Comprehensive two-dimensional HPLC (LC×LC) is a promising method for the separation of such complex samples. LC×LC delivers peak capacities an order of magnitude or more than uni-dimensional LC. In LC×LC, two separation methods are combined in such a manner that the separation in each dimension is retained. As a consequence, the overall peak capacity in a two-dimensional space is (ideally) the product of peak capacities in each dimension. Admittedly, this ideal is only achieved if certain criteria are met, most importantly that both separations are based on different (orthogonal) mechanisms.
Comprehensive 2-D LC may be performed in off-line, on-line or stop-flow modes.19 In on-line LC×LC, fractions from the first dimension are transferred in real time to the second dimension column, usually using multi-port two-position switching valves. This approach provides relatively fast (approximately equal to 1-D HPLC methods) separations, although resolution in the second dimension is sacrificed to meet sampling rate requirements. Off-line LC×LC, by contrast, results in much longer analysis times, but since no constraints are placed on the second dimension separation much higher peak capacities may be obtained. (Stop-flow LC×LC is less often applied, and provides peak capacities and analysis times in between the off-line and on-line approaches).
In fact, the application of on-line LC×LC using a variety of separation modes has been demonstrated for diverse phenolics in wine, beer and herb samples.18,21,22 In our group, we have investigated the off-line coupling of hydrophilic interaction chromatography (HILIC) and RP–LC to improve procyanidin determination.5 Off-line coupling was chosen to obtain the maximal practical peak capacity through optimal second-dimension analyses. In HILIC, procyanidins are separated according to polarity, which effectively provides molecular weight separation based on the number of hydroxyls (individual isomers are co-eluted).23 RP-LC, in contrast, provides isomeric separation according to hydrophobicity — although complete separation of isomers in not possible in a single analysis. HILIC and RP-LC therefore provide highly orthogonal separations. The off-line comprehensive HILIC×RP-LC analysis of cocoa procyanidins is illustrated in Figure 7, where it is evident that the combination of MW (HILIC, y-axis) and isomeric (RP-LC, x-axis) information delivers a complete picture of the oligomeric procyanidin content of this extract. The practical peak capacity for this separation (according to reference 24) is ~3500.
Figure 7: HILICÃRP-LC analysis of cocoa procyanidins. Numbers correspond to degree of polymerisation of procyanidins, A indicates A-type procyanidins. Refer to reference 5 for further details.
The same approach may also be used beneficially for the analysis of other classes of phenolic compounds. This is illustrated in Figure 8 for flavonol glycosides in green tea (practical peak capacity 2500). A very structured elution pattern reminiscent of comprehensive GC is evident from this figure: retention in HILIC increases with the number of attached sugars, while the different flavonol moieties as well as individual mono-, di- and tri-glycosides are resolved in the RP-LC dimension. It may be concluded that HILIC×RP-LC, and LC×LC in general, holds significant promise for the in-depth investigation of complex phenolic fractions such as encountered in numerous natural products.
Figure 8: HILICÃRP-LC analysis of green tea flavonols. Data taken from reference 6. Myr = myricetin, Q = querectin, K = kaempferol, gal = galactoside, glc = glucoside, rha = rhamnoside, xyl = xyloside, rut = rutinoside and hex = unkown hexose.
The analysis of polyphenols represents a severe analytical challenge, and important advances in natural phenolic research hinge on the development of suitable separation strategies. Fortunately, recent developments in HPLC show promise for improving polyphenol analysis.
A summarized comparison of conventional HPLC, UHPLC and/or elevated temperature LC and LC×LC analyses of different phenolic compounds is presented in Table 1. These examples demonstrate that the combination of small dp columns and/or temperature primarily delivers improved peak capacity production rates. This advantage can be practically exploited in one of two ways: (1) to achieve similar peak capacities to conventional HPLC ~3–4× faster (e.g., cocoa procyanidins, green tea flavonols); and (2) to improve the peak capacity for a given analysis time (e.g., blueberry anthocyanins). These techniques show particular promise in improving current routine HPLC methods for polyphenol determination. In addition, application of UHPLC columns operated at elevated temperatures allows improved total peak capacity for complex phenolic fractions (e.g., red wine anthocyanins).
Table 1: Comparison of peak capacities and peak capacity production rates for conventional HPLC, UHPLC, elevated temperature LC and LCÃLC analysis of phenolic compounds in various samples.
For very demanding separations, off-line comprehensive 2-D LC offers greatly improved separation power compared with HPLC (>20× higher peak capacities) at similar peak capacity production rates. The choice of the most suitable separation methodology will therefore depend on the requirements for a particular phenolic sample in terms of speed, instrumental simplicity or resolving power.
It should be noted that state-of-the-art mass spectrometric (MS) techniques such as high resolution MS and tandem MS methods, not addressed in the current contribution, represent a complimentary strategy for improving polyphenol analysis. In fact, MS offers an alternative 'separation' dimension (in terms of m/z ratio), increased selectivity for target analysis and a powerful structural elucidation tool. Combined with advanced chromatographic separation, LC–MS will prove an even more powerful tool for detailed phenolic determination.
Clearly, the judicious application of recent advances in HPLC demonstrably offers significant benefits for phenolic analyses, and shows promise for the in-depth investigation of the phenolic compounds found in numerous natural products — an endeavour which, to date, has been hampered by the lack of suitable separation methods.
The authors gratefully acknowledge Stellenbosch University, the Third World Academy of Science (TWAS, 08-077 RG/CHE/AF/AC) and NRF for financial assistance.
André de Villiers is a senior lecturer at the Department of Chemistry and Polymer Science at Stellenbosch University, South Africa. His research activities include fundamental studies and the practical application of HPLC, CE, GC, MS and sample preparation techniques especially to natural product analysis. He is author or co-author of 24 scientific papers, and the recipient of the 2009 Csaba Horvath Award.
Kathithileni M. Kalili is a first year PhD student at Stellenbosch University. Her research focuses on the development of novel comprehensive multi-dimensional high performance liquid chromatography (HPLC) and electromigration methods for the analysis of complex phenolics.
Mareta Malan is a part-time MSc student at Stellenbosch University. She works for Freeworld Coatings Research Centre in Stellenbosch as an analytical chemist where her research is focused on method development for the analysis of surface coatings.
Jeanine Roodman is currently working as a chemical analyst in a food laboratory. She completed her honours degree in chemistry at Stellenbosch University focused on the application of high temperature liquid chromatography to anthocyanin analysis.
1. E. Haslam, Practical Polyphenolics: From Structure to Molecular Recognition and Physiological Action, (Cambridge University Press, Cambridge, 1998).
2. C. Santos-Buelga and G. Williamson, Methods in Polyphenol Analysis (The Royal Society of Chemistry, Cambridge, 2003).
3. U.D. Neue, J. Chromatogr. A, 1079(1–2), 153–161 (2005).
4. J.M. Davis and J.C. Giddings, Anal. Chem., 55(3), 418–424 (1983).
5. K.M. Kalili and A. de Villiers, J. Chromatogr. A, 1216(35), 6274–6284 (2009).
6. K.M. Kalili and A.de Villiers, J. Sep. Sci., 33(6–7), 853–863 (2010).
7. M.J. Cho et al., J. Sci. Food Agric., 84(13), 1771–1782 (2004).
8. G. Desmet, D. Clicq and P. Gzil, Anal. Chem., 77(13), 4058–4070 (2005).
9. B. Klejdus et al., J. Chromatogr. A, 1195(1–2), 52–59 (2008).
10. M. Schwarz et al., J. Sep. Sci., 32(11), 1782–1790 (2009).
11. F. Lestremau et al., J. Chromatogr. A, 1138(1–2), 120–131 (2007).
12. R. Brouillard and J.-E. Dubois, J. Am. Chem. Soc., 99(5), 1359–1364 (1977).
13. A. de Villiers et al., J. Chromatogr. A, 1216(15), 3270–3279 (2009).
14. A. de Villiers et al., unpublished results (2010).
15. J.D. Thompson and P.W. Carr, Anal. Chem., 74(5), 1017–1023 (2002).
16. G. Guiochon, J. Chromatogr. A, 1168(1–2), 101–168 (2007).
17. F. Gritti et al., J. Chromatogr. A, 1217(10), 1589–1603 (2010).
18. P. Dugo et al., J. Sep. Sci., 31(19), 3297–3308 (2008).
19. R. Mayer et al., J. Agric. Food Chem., 56(16), 6959–6966 (2008).
20. J.N. Fairchild, K. Horváth and G. Guiochon, J. Chromatogr. A, 1216(9), 1363–1371 (2009).
21. P. Cesla, T. Hájek and P. Jandera, J. Chromatogr. A, 1216(16), 3443–3457 (2009).
22. T. Hájek et al., J. Sep. Sci., 31(19), 3309–3328 (2008).
23. A. Yanagida et al., J. Chromatogr. A, 1143(1–2), 153–161 (2007).
24. Z. Liu, D.G. Patterson and M.L. Lee, Anal. Chem., 67(21), 3840–3845 (1995).