OR WAIT 15 SECS
AB SCIEX Application Note
André Schreiber, AB SCIEX, Foster City, California, USA.
This paper describes the use of the AB SCIEX TripleTOF 5600 LC–MS–MS system for targeted and nontargeted screening for pesticide residue in food samples. High resolution and accurate mass MS information was used to quantify compounds and MS–MS spectra were used to identify compounds based on mass spectral library searching.
Fruit and vegetable samples were extracted using a QuEChERS (Quick, Easy, Cheap, Effective, Rugged, and Safe) procedure.1,2 Extracts were diluted 10× to minimize possible matrix effects. UHPLC separation was achieved using a Shimadzu UFLCXR system with a Restek Ultra II Aqueous C18 2.2 μm (100 × 2.1 mm) column and a gradient of water and methanol with ammonium formate buffer. The AB SCIEX TripleTOF 5600 LC–MS–MS equipped with Turbo V source and electrospray ionization (ESI) probe was used. Full scan MS spectra were acquired over a mass range of 100–1000 Da with an accumulation time of 100 ms. Information dependent acquisition (IDA) was used to collect up to 30 MS–MS spectra with an accumulation time of 25 ms for compound identification based on mass spectral library search.
Targeted pesticides were quantified with high selectivity in fruit extracts from extracted ion chromatograms using an extraction window of ±10 mDa. The sensitivity of the developed method was sufficient to allow 10× extract dilution to minimize possible matrix effects, such as ion suppression, and still reach the target limit of quantification for all analytes of less than 10 ppb (μg/kg).
Figure 1: Extracted ion chromatograms, reproducibility and calibration curve (0.05 to 50 ppb of Cyprodinil).
An example of positive findings of pesticide in a red bell pepper sample is shown in Figure 2. The accurate mass MS and MS–MS information was used to identify acephate, carbaryl, clothianidin, dimethoate, imidacloprid, methamidophos, metalaxyl, myclobutanil, omethoate, propamocarb, spinosyn a, spinosyn D, and thiamethoxam with high confidence. MS–MS library searching resulted in FIT values greater than 80%.
Figure 2: Identification of Acephate in a red bell pepper sample using accurate mass and high resolution MS and MSâMS data.
The new AB SCIEX TripleTOF 5600 LC–MS–MS system was used to screen for, quantify and identify pesticide residues in food extracts. High sensitivity of detection allowed dilution of QuEChERS extracts to minimize possible matrix effects. Accurate mass MS–MS spectra were searched against an online LC–MS–MS library of pesticides to confirm the identity of quantified analytes.3
1. M. Anastassiades et al., J. AOAC Int., 86, 412–431 (2003).
2. EN 15662:2007: (2007).
353 Hatch Drive, Foster City, California 94404, USA.
tel. +1 650 638 5800 fax +1 650 638 5884