Phthalates are plasticizers that are added to plastics to make them softer and more flexible. Plastics that contain phthalates are commonly used in the manufacturing of toys. Since phthalates are not permanently bonded to plastics, they can be released from toys through touching, licking, and chewing.
Phthalates are plasticizers that are added to plastics to make them softer and more flexible. Plastics that contain phthalates are commonly used in the manufacturing of toys. Since phthalates are not permanently bonded to plastics, they can be released from toys through touching, licking, and chewing. For the sake of child safety, it is important to qualitatively and quantitatively ascertain the leach-ability of phthalates from plastic toys into saliva. This application note artificially reproduces children licking and chewing toys, followed by a simple extraction to determine phthalate levels leached from toys into artificial saliva.
AS was prepared by mixing 0.18 g xanthan gum, 1.2 g potassium chloride, 0.85 g sodium chloride, 0.05 g magnesium chloride, 0.13 g calcium chloride, and 0.13 g di-potassium hydrogen orthophosphate with 1 L of reagent water (1), and stirred for 4 h.
a) Cut plastic toys into small pieces.
b) Weigh 1 to 2 g of toy samples to a 50 mL centrifuge tube (ECPAHFR50CT).
c) Add 10 mL of AS and two stir bars to the 50 mL tube.
d) Shake for 1 h using a horizontal shaker to mimic children licking and chewing toys.
e) Transfer the AS to a new 50 mL tube, add 10 mL of ethyl acetate and shake for 1 min.
f) Add salts in the Mylar pouch (ECQUUS2-MP) and shake vigorously for 1 min.
g) Centrifuge at 5000 rpm for 5 min.
h) Transfer 1 mL of the supernatant into a 2 mL auto-sampler vial.
i) Add 10 µL of 50 ppm triphenyl phosphate (TPP) as internal standard.
j) The sample is ready for GC–MS analysis.
GC–MS parameters are available upon request.
Table I: Accuracy and precision data (n = 4)*
Matrix matched calibration curves were constructed by analyzing matrix matched standards (0–1000 ng/mL). The responses were linear with correlation coefficient higher than 0.9973. The limit of quantification of this method was 25 ng/mL.
Figure 1: Chromatogram of AS fortified with 200 ng/mL phthalates.
This newly developed method was successfully applied to real toy samples. Diethyl phthalate was found leaching from toy samples into artificial saliva at a concentration of 285 ng/g (RSD = 5.9%, n = 3).
A procedure artificially reproducing the licking and chewing of toys by children is performed using artificial saliva. Phthalates leached from toys are extracted with ethyl acetate and determined by GC–MS. This procedure is simple, fast, and reliable for the determination of phthalates leached from toys into artificial saliva.
A. Preetha and R. Banerjee, "Comparison of Artificial Saliva Substitutes," Trends Biomater. Artif. Organs 18(2), 178–186 (2005).
2731 Bartram Road, Bristol, PA 19007
tel. (800) 385-3153