OR WAIT null SECS
AB Sciex Application Note
The iDQuant Standards Kit for pesticide analysis of certified reference material (ISP Guide 34, IOS/IEC 17025 and ISO 9001:2008) includes 204 pesticides in 10 mixes at a concentration of 100 µg/mL. This kit has been designed to eliminate the need to source pesticides individually, and measure each one manually. This allows users to setup, verify and validate the performance of pesticide screening methods faster.
Many recent developments such as generic extraction procedures, (like QuEChERS), generic LC separation methods, and highly selective and sensitive MS-MS detectors have established LC–MS–MS as a standard technique for the analysis of pesticide residues in food. Hundreds of targeted analytes can be detected in a single analytical run. Software designed for ease-of-use like Cliquid software in conjunction with iMethod applications made the fast adaptation of LC–MS–MS in new routine testing laboratories possible. For a while, data processing was the bottleneck for most laboratories, but the development of fast and automatic data processing and reporting tools sped up the delivery of analytical results.
A major hurdle for each laboratory is method setup, verification and validation. The newly available iDQuant kit can be used for the following tasks:
The iDQuant standards kit was used for method setup and preparation of calibration standards. Fruits and vegetables were extracted using a QuEChERS procedure and diluted 10× with water to optimize chromatographic peak shape and minimize possible matrix effects. LC separation was achieved on a Shimadzu UFLCXR system with a Restek Ultra Aqueous C18 3 µm (100 × 2.1 mm) column and a 15 min gradient of water and methanol with ammonium formate buffer at a flowrate of 0.5 mL/min. The AB Sciex QTRAP 5500 system was operated with Electrospray Ionization. 436 MRM transitions were monitored using the Scheduled MRM algorithm.
The developed LC–MS–MS method was successfully applied to detect pesticides in food samples. Chromatograms were processed using the "multicomponent" query in MultiQuant software to automatically report concentrations above the threshold of 10 µg/kg and to compare MRM ratios for compound identification (Figure 1).
Figure 1: Pesticides above 10 Âµg/kg and with a positive identification based on their MRM ratio following guideline SANCO/10684/2009.
The iDQuant Standards Kit for pesticide analysis was used to successfully setup an LC–MS–MS method to identify and quantify pesticides in fruit and vegetable samples. This was accomplished in less time than was previously possible when a standard kit was either unavailable or had to be created from individual standards.
Copyright 2011 AB Sciex. The trademarks mentioned herein are the property of AB Sciex Pte. Ltd. or their respective owners. AB Sciex is being used under license.
110 Marsh Drive, Foster City, California 94404, USA
tel: +1 877 740 2129 fax: +1 650 627 2803