OR WAIT 15 SECS
KNAUER Application Note
Sample cleanup via solid phase extraction (SPE) and similar methods are time-consuming and cost-intensive procedures. In particular, the purification of complex plant extracts requires special care. Here we introduce a robust and sensitive on-line SPE sample preparation method which was used for the determination of steviol glycosides. Starting with a filtered plant extract, the purification step and steviol glycoside determination requires less than 13 minutes.
The SPE purification and the following determination were performed on a KNAUER PLATINblue HPLC Plus system with two P-1 Pumps (one for analysing and one for transport), Autosampler AS-1, Column Thermostat T-1 with two integrated 6/1 port 1/32" valves, Detector PDA-1, Assistant with two 1/8" solvent selection valves and one separate two way 1/16" valve.
Extraction (Manual): Step 1: Extraction with acetonitrile/water 70:30 (v/v). Step 2: Filtration of the extract through 0.45 µm filter.
Extract purification (Cleanup): Step 1: Injection of 10 µL extract. Step 2: Sample transport to SteviaClean cartridge: cleaning step with acetonitrile/water 90:10 (v/v), 2 min at a flow-rate of 2 mLmin.
Sample Analysis: Switching the SteviaClean cartridge into the determination part of the HPLC system for 30 seconds and starting the PDA acquisition immediately after switching.
HPLC Method: Column: Eurospher 100-5 NH2, 150 × 3 mm. Mobile phase: acetonitrile/water 80:20 (v/v). Flow-rate: 1.0 mL/min. Temperature: 35 °C. Detection: UV, 210 nm.
SteviaClean Cartridge Cleaning and Conditioning (During Sample Analysis): Step 1: After switching back, flushing with water for 4 min at a flow-rate of 3 mL/min. Step 2: Flushing 4 min with acetonitrile/water 90:10 (v/v) at a flow-rate of 3 mL/min and then at 0.5 mL/min flow until the method ends.
As shown in Figure 1 the on-line SPE purification works very fast and robustly with the KNAUER SteviaClean cartridge. A complete purification and determination run can be accomplished within 13 minutes. For standard impurity level plant extracts the SteviaClean cartridge can be used up to 10 times without any penetration of matrix components or losses of stevioside and rebaudioside A. During the subsequent determination, the SteviaClean cartridge was flushed with the listed mobile phases. The slightly shifted retention times between the two runs result from the injection of the purified sample directly from the cartridge. For ten measurements the relative standard deviation of the peak area was 0.4% for stevioside and 0.9% for rebaudioside A. The recovery rate was 107% for stevioside and 99% for rebaudioside A.
Figure 1: Comparison of non-purified extract (blue) and purified extract (red).
This on-line SPE purification method applying KNAUER SteviaClean SPE cartridges in combination with a KNAUER PLATINblue HPLC Plus system is able to reliably purify and determine a stevia plant extract in less than 13 minutes. With the two 6/1 port 1/32" valves installed in the KNAUER Column Thermostat T-1, it is possible to install up to six KNAUER SteviaClean cartridges for the determination of up to 60 samples automatically with an overall run time of approximately 15 hours. Therefore it is a very suitable method for high throughput analysis.
1. Steviol Glycosides, Prepared at the 73rd JECFA (2010), published in FAO JECFA Monographs, 10 (2010).
Wissenschaftliche Gerätebau Dr. Ing. Herbert Knauer GmbH
Hegauer Weg 38, 14163 Berlin, Germany
tel: +49 30 809727 0 fax: +49 30 801501 0