CONTINUE TO SITE >

OR WAIT 15 SECS

- About
- Advertise
- Contact Us
- Editorial Staff
- Editorial Advisory Board
- Ethics Policy
- Do Not Sell My Personal Information
- Privacy Policy
- Terms and Conditions

© 2020 MJH Life Sciences^{™} and Chromatography Online. All rights reserved.

January 1, 2019

*The elution mechanisms for gradient and isocratic HPLC are fundamentally different from each other. Here is what you should understand about both.*

The elution mechanism in gradient high- performance liquid chromatography (HPLC) is fundamentally different to that of the isocratic mode. In gradient HPLC, as the elution strength is initially low, analytes may be stationary within the column, partitioned fully into the stationary phase. As the elution strength increases, the analytes begin to migrate through the column with increasing speed, and, depending upon the gradient profile and column length, may be travelling at the same velocity as the mobile (that is, it does not partition into the stationary phase at all) prior to elution. This contrasts with isocratic HPLC, in which the analyte will move with constant speed through the column, as the elution strength of the mobile phase does not change.

While retention behavior in isocratic HPLC is described using the retention factor (*k*) in gradient HPLC, this is not appropriate, as the retention factor is continuously varying and we must use a new relationship to describe the retention behavior, which is shown in Equation 1:

*t _{g}* is the gradient time in minutes,

*S* is a shape selectivity factor which can be estimated by 0.25 M_{w}^{0.5} (for analytes < 1000 Da a value of 5 is typically used for *S*).

So for the following separation:

- Column: C18 150 × 4.6 mm, 5 µm

- Flow rate: 1.5 mL/min

- Gradient: 20 to 65% acetonitrile (0.1% formic acid) in 7 minutes

*k** would be calculated as follows (equation 3):

For a 'good' method, we would ideally want *k** to lie in the range 2 to 10, and, for an "acceptable" method, *k* *should certainly be in the range 1 to 20.

If *k** is too low, then we risk interference from other sample components or analytes, as the analyte does not have enough affinity for the stationary phase to differentially partition away from other sample components. When *k** is too high, the analysis time is unnecessarily long.

Equation 1 also highlights the important factors which can be easily varied to optimize retention (and therefore selectivity and resolution) in gradient HPLC as the initial and final composition of the gradient (as defined by Δφ), the gradient steepness (as defined by gradient time (*t _{g}*) assuming Δφ remains unchanged) and the eluent flow rate. The column dimensions can also be altered to effect changes in

Of these, it is perhaps most convenient to alter the gradient steepness and we can rearrange equation 1 in a more usable form to allow us to make a prediction of the optimum value of *t _{g}* for a particular separation as given in equations 4 through 6:

In this way, we can predict the optimum gradient time based on the other parameters within the method. In order to generate an "ideal" *k** value of five, in this case a gradient value 12.75 mins would have produced a more ideal separation than the original time of seven minutes.

We may also use another very useful relationship which defines an adjusted gradient steepness (*G _{s}*), and which can be used to maintain analyte selectivity when other factors in method are adjusted as in equation 7:

So, for our original example (gradient time of 12.75 mins), if we wanted to change the column dimensions to 100 × 2.1 mm (with a corresponding *V _{m}* value of 0.232), by maintaining a constant value for

By altering the flow rate to 0.6 mL/min (a sensible flow rate for this new column dimension), we can see that a gradient time of 4.4 mins will produce a theoretically equivalent separation.