OR WAIT null SECS
Tony Taylor is Group Technical Director of Crawford Scientific Group and CHROMacademy. His background is in pharmaceutical R&D and polymer chemistry, but he has spent the past 20 years in training and consulting, working with Crawford Scientific Group clients to ensure they attain the very best analytical science possible. He has trained and consulted with thousands of analytical chemists globally and is passionate about professional development in separation science, developing CHROMacademy as a means to provide high-quality online education to analytical chemists. His current research interests include HPLC column selectivity codification, advanced automated sample preparation, and LCâMS and GCâMS for materials characterization, especially in the field of extractables and leachables analysis.
All sample preparation adds the potential for error to the analytical protocol, so choosing less-complicated techniques will help to reduce inherent error.
There are many reasons for choosing to implement sample preparation, but most fall under the three general aims shown in Table I.
In many cases the type of sample preparation used will depend on the analyte type, the sample matrix, or the instrument technique that will be used for the analysis. In other words, various industries and applications are suited to certain types of sample preparation - some of these have been outlined in Figure 1, alongside the relative specificity of some common sample preparation techniques.
Figure 1: Description of the various commonly used sample preparation techniques. Note that the list of typical applications is for illustrative purposes only and is not meant to be exhaustive.
All sample preparation adds the potential for error to the analytical protocol, so choosing less-complicated techniques will help to reduce inherent error. The use of preconcentration with an inert gas stream for volatile samples or extracts can add significant error to the method.
Knowing the target analyte concentration will help to inform the correct choice of sample preparation technique. The attainable limit of detection (LOD) or limit of quantitation (LOQ) will be directly related to the specificity of the sample preparation technique. It is this parameter that needs to be optimized, as far as is possible, to achieve the cleanest extracts and the lowest limits of detection.
Specificity in sample preparation is optimized when any differences in the chemistry or physicochemical properties of the analyte molecules and the sample matrix are exploited. Strategic ways in which selectivity might be optimized include
For more information on applications and considerations such as extraction method capacity and practical implementation see the CHROMacademy tutorial.