OR WAIT null SECS
© 2023 MJH Life Sciences™ and Chromatography Online. All rights reserved.
LCGC Asia Pacific
PerkinElmer Application Note
P-hydroxybenzoic acid esters, also called parabens, are widely used as antimicrobial preservatives in cosmetic products. Parabens have a potential of endocrine effect — as a result, their use is limited in EU countries to a maximum concentration of 0.4% when used individually and 0.8% as a mixture. In the US, it is against the law to market products with ingredients that may cause injuries under label conditions, thus parabens are typically used at levels between 0.01 to 0.3%. This application note will present a fast, sensitive and reliable UHPLC determination of six common parabens.
This application will discuss throughput improvements and reductions in solvent consumption compared to a conventional HPLC analysis of parabens. In addition, method conditions and performance data will be presented.
The PerkinElmer Flexar FX-15 with Flexar UV/vis detector provided the UHPLC platform for this application. The separation was completed on a PerkinElmer Brownlee Analytical C18, 1.9 µm 50 mm × 2.1 mm column. The run time was approximately 3 min with a back pressure of 8100 psi (559 bar).
The separation was characterized and system calibrated with a mixture of parabens diluted from neat material. Repeatability was studied with six injections of the working standard (1200 ng/mL). Linearity was determined across the range of 30–1200 ng/mL. Samples of face hydrating lotion and cocoa butter body lotion were tested.
A conventional HPLC method would take approximately 16 minutes to resolve the parabens discussed here. This method would use approximately 24 mL of solvent.
Compared with conventional HPLC methodology, the UHPLC run time was dramatically shortened to about 3 min, the resolution of analyte peaks and sensitivity of the determination were improved and the total solvent usage was reduced to 2.1 mL per injection.
A representative chromatogram from standard solution analysis under UHPLC conditions is presented in Figure 1.
Figure 1: Example chromatogram from the analysis of a standard solution of common parabens with UHPLC and a 1.9 Âµm particle size column.
The method performance was outstanding. The linearity of the analysis achieved an average r2 value of 0.998. The average precision was less than 1% RSD, ranging from 0.9 to 0.2%.
The application of UHPLC to the analysis of common antimicrobials in cosmetics has resulted in a 13 min or 80% reduction in run time as well as a reduction of solvent usage of 22 mL, or about 90%. The PerkinElmer Flexar FX-15 UHPLC system and Brownlee Analytical C18, 1.9 µm 50 × 2.1 mm column resolved all the six parabens in about three minutes. The method was shown to be linear and precise. When applied to hand and body lotions, parabens were detected and their amounts complied with the EU and US Cosmetic recommendations.
1. Robert Golden, Jay Gandy and Guenter Vollmer, Critical Review in Toxicology, 35, 435–458 (2005).
2. European Scientific Committee on Consumer Product (SCCP), SCCP/0873/05.
3. Federal Food, Drug and Cosmetic Act (FD&C Act) Cosmetic Ingredient Review (CIR).
940 Winter Street, Waltham, Massachusetts 02451 USA
tel. +1 800 762 4000 or +1 203 925 4602
fax +1 203 944 4904